The use of medical devices containing highly criticized phthalates including di(2-ethylhexyl) phthalate (DEHP) has been challenged by European directive 2007/47/CE, put into effect in March 2010. New plasticizers are now being used to soften PVC in medical devices: trioctyltrimellitate (TOTM), di-isononylcyclohexan-1,2-dicarboxilate (DINCH) and di(2-ethylhexyl) terephthalate (DEHT). To quantify DEHP in nine DEHP-free medical devices made of PVC softened by alternative plasticizers, high performance liquid chromatography analysis with ultraviolet detection at 220 nm wavelength was achieved. An NMR spectroscopy was performed to confirm DEHP presence. Only two medical devices out of the nine tested were truly without DEHP. One of them showed traces of DEHP exceeding the threshold contamination of 0.1% in plastic mass set by REACH regulations. TOTM plasticizer is still incriminated when polyvinylchloride (PVC) is contaminated with DEHP. Manufacturers must verify the purity of their raw material. not only on PVC, but also an other soft plastics entering into the composition of medical infusion devices. The clinical consequences of exposure to certain levels of DEHP have not been evaluated. A solution could be to use alternative PVC-free materials.
Scheda prodotto non validato
Attenzione! I dati visualizzati non sono stati sottoposti a validazione da parte dell'ateneo
|Titolo:||Experimental study on infusion devices containing polyvinylchloride: To what extent are they di(2-ethylhexyl)phthalate-free?|
|Data di pubblicazione:||2011|
|Appare nelle tipologie:||Articolo|